1.The Royal College of Radiologists, the College of Radiographers and the Institute of Physics and Engineering in Medicine (2017). Magnetic resonance imaging (MRI) equipment, operations and planning in the NHS - Report from the Clinical Imaging Board. London, UK.
Retrieved from www.rcr.ac.uk/sites/default/files/cib_mri_equipment_report.pdf
2. Oticon Medical CI Unique – January 2019 (the Oticon Medical Unique logo Indicates an Oticon Medical unique technology or accessory in the CI industry at the time of this publication)
3. Todt, I., Rademacher, G., Grupe G., Stratmann A., Ernst, A., Mutze S., Mittmann P. (2018). Cochlear implants and 1.5 T MRI scans: the effect of diametrically bipolar magnets and screw fixation on pain. Journal of Otolaryngology - Head and Neck Surgery (2018) 47:11
4. Saadat S. et al, Overuse or underuse of MRI scanners in private radiology centers in Tehran, International Journal of Technology Assessment in Health Care 24(3):277-81 · February 2008
5. Grupe G, Wagner J, Hofmann S, Stratmann A, Mittmann P, Ernst A, Todt I. Prevalence and complications of MRI scans of cochlear implant patients: English version. HNO. 2017 Jan;65(Suppl 1):35-40. doi: 10.1007/s00106-016-0129-7. PMID: 26886493
6. Wagner, F., Wimmer, W., Leidolt, L., Vischer, M., Weder, S., Wiest, R., Mantokoudis, G. and Caversaccio, M.D., 2015. Significant artifact reduction at 1.5 T and 3T MRI by the use of a cochlear implant with removable magnet: an experimental human cadaver study. PloS one, 10(7), p. e0132483